skrf.calibration.calibration.Calibration.__init__¶

Calibration.__init__(measured, ideals, sloppy_input=False, is_reciprocal=True, name=None, self_calibration=False, *args, **kwargs)[source]

Calibration initializer.

Notes

About the order of supplied standards,

If the measured and ideals parameters are lists of Networks and sloppy_input=False, then their elements must align. However, if the measured and ideals are dictionaries, or sloppy_input=True, then we will try to align them for you based on the names of the networks (see func:align_measured_ideals).

You do not want to use this sloppy_input feature if the calibration depends on the standard order (like TRL).

Parameters: measured (list/dict of Network objects) – Raw measurements of the calibration standards. The order must align with the ideals parameter ( or use sloppy_input) ideals (list/dict of Network` objects) – Predicted ideal response of the calibration standards. The order must align with ideals list ( or use sloppy_input) sloppy_input (Boolean.) – Allows ideals and measured lists to be ‘aligned’ based on the network names. is_reciprocal (Boolean) – enables the reciprocity assumption on the calculation of the error_network, which is only relevant for one-port calibrations. name (string) – the name of this calibration instance, like ‘waveguide cal’ this is just for convenience [None]. self_calibration (Boolean) – True if there are less ideals than measurements. Used in self-calibration such as LMR, LRRM, where some of the standards can be unknown. **kwargs (*args,) – stored in self.kwargs, which may be used by sub-classes most likely in run.